RADAR LEVEL TRANSDUCER - TLR
Measure principle: The transducer produces a series of microwave impulses that are sent from radar emitter to measuring level. The reflected microwaves are noticed by spar and sent to electronic devices where a microprocessor appraises signal and identifies the reflected echo coming from measured surface calculating the distance from transducer.
Intelligent sensor: The transducer belongs to the intelligent sensor family as based on a micro-processor chip that execute among other functions: the correct control functioning, data pre-processing, electrical signal A/D conversion etc. This characteristics warranty excellent accuracy, long-term stability and high data reliability.
Laboratory calibration: The sensor can be supplied with a SIAP+MICROS calibration certificate or with calibration certificates issues by external primary metrological services (SIT, Colonetti, ecc).
Housing materials and International standards: The external structure is made by die casting aluminum, while the parts attached by liquid are made by stainless steel AISI316L. The transducer complies with CE EMC standards, it’s protected against over tensions and is fully compliant within the WMO (World Meteorological Organization) standards.
Range: 0,6 ÷ 15 m
Analog output range: 4÷20mA into configurable range
Sensitivity: .5 mm
Accuracy: ± 1 mm
Working temperature: -30 ÷ +80°C
Power supply: 10÷16 Vdc
Max load : 390 ohm
Supply current (mA): 50 mA
Protections: Polarity reverse and transient
Housing: alluminum alloy and AISI 316L
Weight: 4000 g
Dimensions: 143x193x162 mm
This sensor is installed into these stations:
Ancona, Anzio, Bari, Cagliari, Carloforte, Catania, Civitavecchia, Crotone, Gaeta, Genova, Imperia, La Spezia, Lampedusa, Livorno, Marina Di Campo, Messina, Napoli, Ortona, Otranto, Palermo, Palinuro, Ponza, Porto Empedocle, Porto Torres, Ravenna, Reggio Calabria, Salerno, San Benedetto Del Tronto, Sciacca, Taranto, Tremiti, Trieste, Venezia, Vieste