WIND SPEED TRANSDUCER - T006 TVV
Measure principle: the sensing element is constituted by a 3-cup Robinson assembly coupled to an optical encoder disk, a series of infrared emitter-receiver LEDs (Light Emitting Diode) detects the encoder disk movement allowing in this way to calculate the wind speed.
Intelligent sensor: The transducer belongs to the intelligent sensor family as based on a micro-processor chip that execute among other functions: the correct control functioning, data pre-processing, electrical signal A/D conversion etc. This characteristics warranty excellent accuracy, long-term stability and high data reliability.
Laboratory calibration: The sensor can be supplied with a SIAP+MICROS calibration certificate or with calibration certificates issues by external primary metrological services (SIT, Colonetti, ecc).
Housing materials and International standards: The dimensions and materials used warranty high sensibility, low mechanical inertia and a low starting threshold. The transducer complies with CE EMC standards, it’s protected against over tensions and is fully compliant within the WMO (World Meteorological Organization) standards.
Range: 0 ÷ 50 m/s
Resolution: 0,1 m/s
Sensitivity: 0,25 m/s
Conversion constant: 30 Hz/m/s
Accuracy: ±0,5 m/s (0÷20m/s) ; ±1 m/s (>20 m/s)
Transducer: 3 cups with optical transducer
Working temperature: -30 ÷ +60 °C
Power supply: +10 ÷ +16 Vdc
Output resist (ver. TVV-V): 10 ohm
Max load (ver. TVV-I): 390 ohm
Protections: polarity reverse and transient
Housing: Aluminum alloy and stainless steel
Weight: 750 g
Dimensions: ømax = 230 mm ; h = 312 mm
Connector: (IP67) 4 poles male
This sensor is installed into these stations:
Ancona, Anzio, Bari, Cagliari, Carloforte, Catania, Civitavecchia, Crotone, Gaeta, Genova, Imperia, La Spezia, Lampedusa, Livorno, Marina Di Campo, Messina, Napoli, Ortona, Otranto, Palermo, Palinuro, Ponza, Porto Empedocle, Porto Torres, Ravenna, Reggio Calabria, Salerno, San Benedetto Del Tronto, Strombolicchio, Taranto, Trieste, Venezia, Vieste